# Quantitative Fluorimetry

## Introduction

Light emission from atoms or molecules can be used to quantitate the amount of the emitting substance in a sample. The relationship between fluorescence intensity and analyte concentration is:

F = k * QE * P_{o} * (1-10^{[-*b*c]})

where F is the measured fluorescence intensity, k is a geometric instrumental factor, QE is the quantum efficiency (photons emitted/photons absorbed), P_{o} is the radiant power of the excitation source, is the wavelength-dependent molar absorptivity coefficient, b is the path length, and c is the analyte concentration (, b, and c are the same as used in the Beer-Lambert law).

Expanding the above equation in a series and dropping higher terms gives:

F = k * QE * P_{o} * (2.303 * * b * c)

This relationship is valid at low concentrations (<10^{-5} M) and shows that fluorescence intensity is linearly proportional to analyte concentration.

Determining unknown concentrations from the amount of fluorescence that a sample emits requires calibration of a fluorimeter with a standard (to determine K and QE) or by using a working curve.

## Limitations

Many of the limitations of the Beer-Lambert law also affect quantitative fluorimetry. Fluorescence measurements are also susceptible to inner-filter effects. These effects include excessive absorption of the excitation radiation (pre-filter effect) and self-absorption of atomic resonance fluorescence (post-filter effect).

## Specific fluorescence techniques

Copyright © 2000 by Brian M. Tissue, all rights reserved.